Service company, LLC
I send you review on the operation of the instrument milli-ohmmeter «MIKO-7MA», this feedback is compiled on the basis of the annual operation of this instrument in various weather conditions and at various measurement objects.
Milli-ohmmeter «MIKO-7MA» is usually used to measure the transformers parameters, namely to measure DC resistance of windings, as well as for measuring the contact resistance of commutation devices during start-up and commissioning and operation.
During the operation, there were no significant remarks to the instrument except for a very long charging, but this comment is not essential as it was most likely due to the low temperature of the battery after transportation. I believe this instrument is better than the competitor, as measurements are faster, battery charge lasts for a long time, and the instrument works more «Stable».
[Manufacturer note]: According to the operation requirements, which are reflected in the documentation supplied with the instrument: "The process of charging the battery of the instrument MIKO-7M(A), after long-term storage can be long - up to 48 hours ..."
From the improvements I would like to have longer cables, as well as the connectors that are installed on them from the connection of the «Crocodile» clamps would have a more rigid fixation to these clamps.
[Manufacturer note]: SKB EP company manufactures a wide range of test cables of different lengths, types of fixation and construction, which can be found on the website in the section of a specific instrument complete set. In addition, we can always develop a test cable on an order according to the Client's parameters.
Head of the electrical measurement laboratory of LLC «SC»
Branch of IDGC of Center, PJSC - Kurskenergo
In the Western Section of the Diagnostic Service of Rosseti Kurskenergo, PJSC, 2 measuring instruments of SKB EP, LLC are used: PKR-2M instrument for control of OLTC devices of transformers and milli-ohmmeter MIKO-8M. The PKR-2M instrument has been used since 2018, the MIKO-8M instrument has been used since 2019.
These instruments are used for diagnostic of power and measuring transformers with a voltage of 110, 35, 10 kV (MIKO-8M), OLTC devices of power transformers 110, 35 kV (PKR-2M).
In practice, both instruments have proven themselves from the best side: they have a compact, durable body, an intuitive and user-friendly interface. There is an opportunity to select the necessary circuits and groups of connection of windings, create and maintain an archive, including the possibility of entering factory tests, this allows to conduct a primary analysis of diagnostic results directly at the workplace, they have high-quality execution of wires, clamps and there is opportunity to adjust the complete set. In comparison with the previously used instrument assortment, all these significantly reduce time and labor expenditures during equipment diagnostic. Also, we would like to note the availability of software for analyzing and storing diagnostic results on a PC and the opportunity to get the necessary information and (or) advice directly from the instrument developers.
Head of the Western Section of the Diagnostic Service
Kumerautskaya CHP, LLC
We inform you that the instrument PKV/M7 (prod. N2 53, 2012), milli-ohmmeter MIKO-8M (prod. N2 035J, 2018) manufactured by SKB EP, LLC (Specialized Design Bureau of Electrotechnical Instrumentation, LLC) are in operation at Kumerautskaya CHP, LLC.
With the PKV/M7 instrument, diagnostic and checking of electrical characteristics of high-voltage circuit breakers are carried out at the station, and the MIKO-8M milli-ohmmeter is used to measure DC resistances of windings and coils of electrical equipment, contact resistances of high-voltage circuit breakers and electrical equipment.
Owing to the fact that the instruments are realized on the basis of microprocessor technologies, it has become much more convenient and efficient to carry out measurements, the accuracy of measurement results and safety have increased, and archiving has become possible.
Deputy Chief Engineer - Head of the Technical Department of Kumertautskaya CHP, LLC
The Vologda Branch of Interregional Distribution Grid Company of North-West, PJSC
We thank SKB EP, LLC for effective and mutually beneficial cooperation lasted for more than 10 years. SKB EP, LLC has established itself as a reliable partner and manufacturer of instruments and complete sets during the time of cooperation.
I would like to inform you that the production department of the Vologda Electric Networks uses such instruments manufactured by SKB EP, LLC as PKV/M7, MIKO-1 and manipulating rod.
These instruments are used for diagnostic of the high-voltage circuit breakers both produced in our country (VMT-110, MKP-110, VT-35, S-35) and abroad (LTB-145), and for measuring contact resistance of various switching devices. During the time of use, these instruments have proven themselves from the best side. Use of the PKV/M7 allowed realising non-demountable detection of the circuit breakers internal defects.
The instrument is very simple and easy to operate, it allows to control the main characteristics of high-voltage circuit breakers. During repair, the main malfunction that were detected by the instrument was time difference between the contacts closing. When analyzing the speed characteristics, speed dips were detected, that indicated the presence of jamming, mashing in the mechanism and necessity of repairing.
Before this instrument was gotten, all the characteristics of the circuit breakers had to be checked with mandatory disassembly, that increased the repair time. Use of the PKV/M7 instrument with a manipulating rod can significantly reduce time for checking of the characteristics of both column and tank circuit breakers of 110 kV, and also eliminates the need for scaffolding or the use of lifting mechanism to conduct these works.
Our company is engaged in installation, start-up and commissioning, as well as repair work of electrical equipment.
In the course of the work, it became necessary to purchase the PKR-2 instrument, which in turn significantly simplified (made easier) reparation of the drives of the OLTC regulators of power transformers.
The instrument has a sufficiently informative screen, easy connection, it provides accurate measurements, and it is reliable and compact.
Also, the service functions that your company provides for the manufactured equipment are important.
Our company also used other instruments that your company produced. We can confidently recommend them for purchase by other companies.
Head of the electrical measurement laboratory of Electroshield-Ufa LLC